We have ADQ412AC-4G-MTCA digitizer cards for the experiment crates (Exp1, Exp2 and FL2Exp1):
ADQ412_datasheet.pdf ( new version, sampling rates table)
The cards are connected via Patch panels to the Experimental endstations and can be operated with a jddd panel:
|Impedance AC||50 OHM|
|Input voltage range||800 mV pp!!!!!!!!!!!!!!!!|
|Digitizer resolution||12 bit|
4 CHANNELS MODE
|Sampling rate||2 *||GSPS|
2 CHANNELS MODE
|Sampling rate||4 *||GSPS|
* note that the sample rate is NOT locked to the FLASH repetition rate ! Thus there is a not integer number of samples between FLASH pulses. The sample rate also differs slightly from ADC card to ADC card. The rough spacing is 1993.846 samples between 2 pulses at 1 MHz for details ask the local contact.
The HDF5 names for the ADC traces are depending on the beamline :
/FL1/Experiment/PG/ADQ412 GHz ADC/CH00/TD
/FL1/Experiment/PG/ADQ412 GHz ADC/CH01/TD
/FL1/Experiment/PG/ADQ412 GHz ADC/CH02/TD
/FL1/Experiment/PG/ADQ412 GHz ADC/CH03/TD
/FL1/Experiment/BL1/ADQ412 GHz ADC/CH00/TD
/FL1/Experiment/BL1/ADQ412 GHz ADC/CH01/TD
/FL1/Experiment/BL1/ADQ412 GHz ADC/CH02/TD
/FL1/Experiment/BL1/ADQ412 GHz ADC/CH03/TD
/FL1/Experiment/BL2/ADQ412 GHz ADC/CH00/TD
/FL1/Experiment/BL2/ADQ412 GHz ADC/CH01/TD
/FL1/Experiment/BL3/ADQ412 GHz ADC/CH02/TD
/FL1/Experiment/BL3/ADQ412 GHz ADC/CH03/TD
and at FLASH2
/FL2/Experiment/MTCA-EXP1/ADQ412 GHz ADC/CH00/TD
/FL2/Experiment/MTCA-EXP1/ADQ412 GHz ADC/CH01/TD
/FL2/Experiment/MTCA-EXP1/ADQ412 GHz ADC/CH02/TD
/FL2/Experiment/MTCA-EXP1/ADQ412 GHz ADC/CH03/TD
DOOCS prop :
FLASH.FEL/ADC.ADQ.PG/EXP1.CH00/CH00.TD or CH00.DAQ.TD
CH00.TD is the full ADC trace as it is sampled ( typically several 100.000 samples per pulse train) while the
CH00.DAQ.TD trace only has the number of samples which are sent to the DAQ OR if grouping is activated the
CH00.DAQ.TD contains only the grouped spectra. To read the ADC trace with an online analysis program the
CH00.DAQ.TD is preferable to use ...
in addition there are also additional parameters saved like:
There is one in each of the MTCAs in the hall: MTCA-EXP1 at PG/BL1, the other at BL2 and BL3.
They are 16 bit, 10 channel, 125 MS/s ADCs.
50 Ohm input impedance, -1 V,...,+1 V default input range, analog signals can be routed to AC and DC input stage. The coupling is DC via op-amp (switching to AC transformer involves resoldering of SMD solder bridges).
Here is a trace of the first signal, a 1 MHz trigger connected from the x2timer board in the same MTCA:
FS-LA (Falko Peters) programmed a pulse detection server that automatically detects peaks in the signal and integrates the samples around the peak.
Things to set:
jddd server panel Definition of the parameters