- the relevant data is the Arrivaltime FLASH.SDIAG/BAM/4DBC3/LOW_CHARGE_ARRIVAL_TIME
- besides the arivaltime from FLASH1 there is also the FLASH2/3 electron arrival time saved. The BAM data is saved for the complete RF pulse. First bunches are from FLASH1 then there is a gap for switching and then there is a second part for FLASH2 (starting at the FLASH2 start time (recorded in DAQ as
/FL2/Timing/start time flash2))
- There are now (since March 2017) LOW and HIGH charge channels. for now the LOW_CHARGE Channel is the relavant one.
- bigger numbers indicate later arrivaltime of the electrons
- the arrivaltime sould be within -20 ps and +20 ps . otherwise there might be a problem ...
- There is a BAM status parameter: FLASH.SDIAG/BAM/4DBC3/BAM_STATUS.1 (the 1 indicates FLASH1)
- bit 0 (value: 1) = data is valid,
- bit 1 (value: 2) = at least 1 bunch is present,
- bit 2 (value: 4) = calibration is on,
- bit 3 (value: 8) = internal feedback is on,
- bit 4 (value: 16) = tuning mode is on