(Okt 2017 Frank-Uwe)
After months of not in use all piezo motors run to the very end
a) disable piezo motors
In the manual of Nanos (Nanos_Doku_02b.pdf page7) a disabling of the motor driver is mentioned. I will ask Pontus, if that is integrated.
Does not help - if there is a current off or a restart of the tango device server, the ParkingActive is off as well as the PIDActive. P11 has similar problems.
Feb 2018 New Crystal holder for all 4 crystals.
Prealignement at QS department. Results see below
runs very fast to 0, but continues
then to move
in negatvie direction. stopped by -100.000
with new homeposition 0. Gap now 2,6mm
same problem / jumps immediately to 0, but
this time it stays there.
homing did not act properly - go tot jive, right mouse click
and restart device